|Registry of biomedical companies:
   [A] [B] [C] [D] [E] [F] [G] [H] [I] [J] [K] [L] [M] [N] [O] [P] [Q] [R] [S] [T] [U] [V] [W] [X] [Y] [Z] 589 active entries
-/ 1440 Barberry Drive Port Coquitlam BC V3B1G3 CA
Toll free: 1-778-554-6691
We are in the process of consulting future licensing manufacturers for subsidiaries in MNL (Philippines), Boston MA USA and Sydney AUSTRALIA.
Design of our Digital Densitometer Scanner provides two (2) formats:
(1) DM = desktop mounted for lab bench
(2) M = mobile dedicated device for biofermentation design, scale-up and prototyping and routine monitoring for production
I LASER's Defn. = light amplified by stimulated emission of radiation
-Laser is a photonic pulsed laser with focused, sensitive, specific and multiple readings and is accurate
II Components of Device
Ar(g) Laser (pink) 45 mW with galvanometer rotors (with transistors, ICs) using deflecting and splitting mirrors (reference & analytical beams) that is top mounted (desk mounted) or front stack (vertical mounted); the beam is split into 2- the reference and analytical beams by a grating and configured via microappliances driven by IC chips using mirrors to project to the electronic detector while the other acros 2 before and after confocal lenses to focus through the column and then focus diffusion of the analytical beam to the electronic detector.
Reader or plate holder with chassis / body of smoked resin / glass on galvanometer rotors ( with transistor ICs)
Detector is light (lambda) activatable using transistored photovoltaic cells; travels along galvonometer rotors (with transistor ICs) that is bottom mounted (desk mounted) or back stacked (vertical mounted)
WiFi operated with laptop unit using a DM or M (in-house unit as a dedicated device)
III READability of LASER Specifications:
Issue #1: Can analyte band be intense enough to be measured, viz., if it is too strong past 'the saturation point' of absorptivity (epsilon)
Issue #2: The duration of time sampling (pulsed) allows for very fast instantaneous readings: can pick up lots of information due to sensitivity; and sensitivity is excellent without reaching saturation of aborptivity (epsilon); the latter being the point beyond which there is no more significant increase in detectable absorptivity (epsilon) with respect to the reference beam
IV MODEL of a Digital Laser Densitometer (Multiplex): GE Amersham Molecular Dynamics Typhoon 9410 Molecular Imager
(Google this title or entry for the link)
WE believe in: (1) Sustainable technology of PAGE electrophoresis such as SDS, and in this case, DT due to its continuing cost-effectiveness, sensitivity, reproducibility and accuracy comparable and if not better than other techniques such as HPLC; (2) the reagent demonstrated here is unique to our proprietarship with no other known absorption tag known (extant) in the market and (3) the time savings advantage that is indeed problematic in this technique cf. to Coomassie blue due to staining/ destaining like SDS; i. e. the science literature suggests a unique caveat we are carving as we speak.
D. A. Flores (c) 2017 Skye Blue Publications. Port Coquitlam, BC V3B1G3 CANADA
Last update of this entry: November 30, 2017
Bookmark and Share this page (what is this?)
Social bookmarking allows users to save and categorise a personal collection of bookmarks and share them with others. This is different to using your own browser bookmarks which are available using the menus within your web browser.
Use the links below to share this article on the social bookmarking site of your choice.
Read more about social bookmarking at Wikipedia - Social Bookmarking